Please use this identifier to cite or link to this item:
- A new tuning fork with different vibration frequencies as an aid to bronchopulmonary hygiene physiotherapy
- Alcântara, José Roberto de
- Manzano, Roberta Munhoz
- Gonçalves, Maicon Gabriel
- Santos, Rodrigo Leonel dos
- Albino, Daniel Donaire
- Apostólico, Nadua
- Fonseca, Nina Teixeira
- Urbano, Jéssica Julioti
- Daibem, Célio Guilherme Lombardi
- Donner, Claudio F.
- Gimenes, Camila
- Ambrozin, Alexandre Ricardo Pepe
- Oliveira, Luis Vicente Franco de
- Faculdades Integradas de Bauru (FIB)
- Universidade Paulista (UNIP)
- Universidade Nove de Julho (UNINOVE)
- Multidisciplinary and Rehabilitation Outpatient Clinic
- Universidade Estadual Paulista (UNESP)
- Background: The main function of the mucociliary system is the removal of particles or substances that are potentially harmful to the respiratory tract. The tuning fork therapeutic for the purpose of bronchial hygiene has still not been described in the literature. The optimal vibration frequency to mobilize secretions is widely debated and varies between 3 and 25 Hz. It is expected that a tuning fork is able to generate vibrations in the thorax, facilitating bronchial hygiene. The aim of the present study is to develop tuning forks with different frequencies, for use in bronchopulmonary hygiene therapy. Methods: The first tuning fork was made with a fixed frequency of 25 Hz and it was recorded in the Brazilian institution of patent registration. This device generated a frequency of 25 Hz and had a weight of 521 g, with dimensions of 600 mm in total length. The device is characterized by a bottom end containing a transducer with a diameter of 62 mm and a thickness of 5/16 mm (8''), a rod removable 148 mm, fork length of 362 mm and an extension at the upper end of sinuous shape bilaterally.The tuning forks must be applied at an angle of 90° directly on the chest wall of the patient after pulmonary auscultation for location of secretions. The tuning fork is activated by squeezing the tips of the extensions together and releasing them in a sudden movement. Results: This study shows the result of the development of others three tuning forks of different dimensions to generate different frequencies. Each equipment reaches a fixed frequency preset of 12, 15 and 20 Hz measured by digital oscilloscope. Conclusions: The tuning fork models developed in this study generated different frequencies proposed by the scientific literature as effective in the mobilization of pulmonary secretions.
- Multidisciplinary Respiratory Medicine, v. 9, n. 41, p. 1, 2014.
- Acesso aberto
There are no files associated with this item.
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.